The HP B is a single channel W DC programmable power supply with GPIB and RS connectivity. It is part of the HP x series. Agilent B/B/B/B System DC Power. Supplies All references to HP-IB have been changed to GPIB. A note about. I just picked up two HP B power supplies as I read somewhere quickly that it not only can be programmed via GPIB to set voltage/current.
|Published (Last):||13 April 2017|
|PDF File Size:||20.41 Mb|
|ePub File Size:||2.67 Mb|
|Price:||Free* [*Free Regsitration Required]|
Please enter a question. Unless otherwisenoted, both units will be referred to by the description “dc source” throughout this manual. If this occurs, turn the power offand then back on to see if the error persists. It also made the fan run quieter.
Current Measurement Low Range Action7.
Release the key after you press it. CC The dc source output is in constant-current mode.
On the Function keypad, press Protect. OCP The overcurrent protection state is enabled. Alexa Actionable Analytics for the Web.
HP (Hewlett-Packard) 6632B Power Supply User Manual
Measurement samples may notcoincide with the actual maximumor minimum point of the waveform. Page 75 Manual UpdatesThe following updates have been made to this manual since the November printingindicated on the Printing History page. Option is not available on HP B units. A – SpecificationsTable A When set to Fast mode, this switch disconnects the output capacitor that is located inside theunit. CopyleftAll Wrongs Reserved.
CError MessagesError Number ListThis appendix gives the error numbers and descriptions that are returned by the dc source. East Dane Designer Men’s Fashion. Amazon Restaurants Food delivery from local restaurants. Or at least you can get close.
VoltagePress this key to access the voltage menu. Installation – 3Under normal operation, the dc source senses the output voltage at the output terminals on the backof the unit.
HP B : – Hewlett-Packard Power Supply User Manual
Minimum clearances for bench operation are 1 inch 66322b mm along the sides. The following table lists the errors that are associated with SCPI syntax errors and interfaceproblems. ASpecificationsSpecificationsTable A-1 lists the specifications of the dc source. Many people prefer to read the documents not on the screen, but in the printed version.
Amazon Renewed Refurbished products with a warranty. There’s a problem loading this menu right now. All programmable settings are saved.
Verification and Calibration – BTable B A – SpecificationsSupplemental CharacteristicsTable A-2 lists the supplemental characteristics, which are not warranted but are descriptions oftypical performance determined either by design or type testing. Therefore, the total acquisition time for a single front panelmeasurement is about milliseconds. I have two such supplies and they are really cool.
CV The dc source output is in 66632b mode.
TICS International Ltd., UK
Verification and Calibration – BTable B- 3. In this mode the portfunctions as a remote inhibit input with a discrete fault indicator output signal. Page 3 Safety SummaryThe following general safety precautions must be observed during all phases of operation of this instrument.
Component replacement and internal adjustments must be made only byqualified service personnel. Ebay buyers of HP supplies take note: How you did the voltage calibration? The dc source must be in set to Local mode to use the front panel controls. The next problem was the fact that the front panel meter was a little off from what I was measuring at the output.