View and Download Gefran user manual online. SINGLE DISPLAY TEMPERATURE CONTROLLER WITH UNIVERSAL INPUT. Temperature Controller. Gefran Pdf User Manuals. View online or download Gefran User Manual . Gefran PID Temperature Controller, 48 x 48 (1/16 DIN)mm, 3 Output Relay, Gefran. Main Product. Technical Reference. / Instruction Manual.
|Published (Last):||8 April 2011|
|PDF File Size:||14.65 Mb|
|ePub File Size:||8.78 Mb|
|Price:||Free* [*Free Regsitration Required]|
Styled for laboratory or industrial applications CAL P has an impact resistant smoked fascia gefran 400 manual rubberised Manuap return to normal functioning, press key F for 5 secs. It Integral heating time 0, Send Cart in an Email Done!
Gefran User Manual – Industores
You can change your cookie settings by reading our manuaal policy. Model can have an auxiliary input for a 50mAac current transformer to monitor gefran 400 manual load current. Default Custom 22 tr 0 tyP Type of sensor, signal and main input scale 0 dP. To connect the relay outputs contactors, solenoid valves, motors, fans, etc.
Resistors must be at least 2W ; gsfran a 1N diode in parallel with gefran 400 manual coil of inductive loads that operate in DC.
The calculated parameters are saved. If an oscillation persists, increase the proportional band. The variable for example, temperature must be that assumed at zero power room temperature. This provides a very low gefran 400 manual current, suitable for an electronic measurement circuit.
PID parameters are calculated by measuring overshoot and the time needed to reach 40. S Digital filter gefran 400 manual main input 0, Fit a 1N diode in gefraj with the coil of the inductive loads that operate in Direct Current.
Page of 12 Go.
gefran 400 Manuals
It will be similar to that in the figure: To semplify the configuration even further, a programming kit is available for PC, which includes a menu driven configuration program for Windows and the necessary cable to connect the instrument see technical data cod. If the Integral Time value is too long Weak maual actiondeviation between the controlled variable and the setpoint may persist. It is essential that the plant has a good earth connection: The controller supplies maximum power until an gefran 400 manual value between starting value and setpoint is reached, after which it zeros power.
The same time has to run after one-shot. Gefran 400 manual secondary current may be detected as voltage on a resistor. Go to your whishlist. When the electronic instrument power gefran 400 manual is strongly disturbed by the commutation of transistor or power units or motors, an isolation getran should be used for the controllers only, earthing the screen.
It is activated via the Stu parameter as can be seen on the table, gefran 400 manual values to be set depend on whether Self-tuning or Soft-start is enabled.
The direct HB alarm trips if the value of the current transformer manuap is below the set limit for Hb. Do not clean the case with hydrocarbon-based solvents Petrol, Trichlorethylene, etc.
Function block formatting allows gefran 400 manual simplified parameter programming.
How to switch the unit ON: The West N 040 is a competitively priced, gefran 400 manual level product which features a simplified operator interface and automatic self-tuning on process start-up. The Wilcoxon accelerometers are of great value to a wide range of applications machinery such Before proceeding with installation read the following warnings gefran 400 manual Short Operating Instruction for temperature regulator We deliver our temperature regulator in its basic adjustment for the control of heaters.
It is interrupted if the set-point is changed, gefarn automatically resumes with a constant set-point. For more information click here.
Manua instrument must NOT gefran 400 manual installed in flammable or explosive environments; it may be connected to equipment operating in such atmospheres only by means of appropriate and adequate types of interface, conforming to the applicable safety standards. It Integral cooling time 0, Den Schutz stellen Sie wie folgt ein: X Maximum limit heating power 0, S Gefran 400 manual limit of main gefran 400 manual scale min A configurable personal software protection code password protection can be used to restrict the levels of editing and displaying the configuration parameters.
Deviation indicator On if deviation is between 0. The product does not contain any of the restricted substances in concentrations and applications banned by the Directive, and for components, the product is capable of being worked on at the higher temperatures required by lead—free soldering.
F If possible, to optimize parameters, change the setpoint and check temporary response.
PL coldand assesses the effects in timed overshoot. Turn off the instrument gefran 400 manual. PID parameters are calculated by measuring overshoot and the Process time needed to reach peak. Send as email Print page. To print the manual completely, please, download it. Set the setpoint to the required value 4.
C Switch to automatic and observe the behavior of the variable.