This standard is issued under the fixed designation C; the number Conversion and rounding given in Standard IEEE/ASTM SI. The intention of this article is to clarify the specifications listed in ASTM C, Standard Specification for Mixing Rooms, Moist Cabinets, Moist. ASTM C Mixing Rooms, Moist Cabinets, Moist Rooms, and Water Storage. Tanks Used in the Testing of Hydraulic Cements and Concretes.
|Published (Last):||17 February 2010|
|PDF File Size:||3.32 Mb|
|ePub File Size:||18.40 Mb|
|Price:||Free* [*Free Regsitration Required]|
Learn how to be an AASHTO lab 24/7
ast In order to maintain an extremely humid environment, you must have a good water supply and humidifying equipment, and the temperature must be maintained at It is informational only and not an official part of the standard; the full text of the standard itself must be referred to for its use and application.
Also consider all the man hours lost to constant maintenance and quick-fixes, not to mention having to spend time writing corrective action reports for your AASHTO re: Some of the drawbacks include the asfm drying out and having to be replaced often, and the temperature not reading correctly.
C11 system used in most grocery store produce sections works very well for this purpose. Dry conditions, low temperatures, and abundant fresh water leaching can all lead to low strength specimens.
ASTM C Standard
It is best to have them inside and against an interior wall. Moist rooms can be expensive to build and maintain, but they also hold more specimens, which are easier to organize and locate when placed on shelves.
This article can be utilized as a general guideline for curing materials that include Portland awtm concrete, masonry mortar, masonry cement, concrete beams, grout, and hydraulic cement see Table 1. If the calcium hydroxide is disappearing when stirred, the water is not saturated.
The gold standard for laboratory accreditation in the construction industry.
Since there seems to be a lot of confusion and misunderstanding regarding curing facilities and their requirements, I decided to write an article about them. Link to Active This link will always route to the current Active version of the standard.
Pre-fabricated curing rooms are not on the market, so most labs make their own out of readily available materials. ICS Number Code If you have more than one tank, you have two options for configuration. Some of the most common errors include: In both cases, extreme care must be taken to prevent water from dripping onto the stored specimens, since lime can leach out and lower the strength.
Trying to cut corners on curing can produce erroneous test results.
ASTM C, Test equipment Controls
Whichever option you choose, you must be capable of heating or cooling the system, regardless of your geography, even if the laboratory is located in a region with a fairly consistent climate. First, heavily consider the placement of your moist room or tanks. C151 Documents astk separately The documents listed below are referenced within the subject standard but are not provided as part of the standard. Take the time to install insulation, piping, replaceable misters, heating and cooling units, circulators, and asstm recording devices as applicable.
It is important to choose temperature charts that are easily read and do not have a scale so small that the pen or marker line width exceeds the scale lines. The temperature of the solution in the tanks must also be monitored.
Thanks for reaching out to us.
If you have to install a moist room against an exterior wall, try to give at least 6 inches of dead air space. Saturated C cure tanks have a considerable amount of excess calcium hydroxide sitting in the bottom of the tank. Tanks near drafty bay doors will also wreak havoc on any kind of temperature control you think you have.
While it is nice to be able to have all of the temperature records on a computer, CCRL inspectors report that laboratories using these data loggers often asym to download and evaluate the data. Accidentally draining and overflowing a tank can occur if there is an equipment malfunction.
Tanks compliant with C will also meet the requirements of C No other units of measurement are included in this standard. Depending on the type of material you are working with, you should refer to the applicable specification or standard to determine whether you c51 to maintain tanks or a curing room, or even a tank within a curing room. The specified relative humidity in these systems shall be maintained by the use of one or more fog sprays, water sprays, or curtains of water on the inner walls.
The system shall be equipped with a temperature recorder and a reference temperature measuring device that are to be placed practically near to each other.
You may use either a chart recorder or a digital recorder. Digital temperature recorders or data loggers are gaining popularity.